This unit has oxygen failure alarm and N2O cut-off
system. When the pressure of oxygen falls down
less than 1 1/2 L.P.M. The machine will not
work N2O Gas will cut down. Oxygen failure alarm
automatic will be on.
It is lengthy alarm you have
to close N2O cylinder to stop alarm.
Tubular, rigid trolley (also
Powder coated mils steel version is available)
Gas specific (pin indexed)
Pressure gauges (two nos
each for Oxygen and Nitrous oxide) conveniently
positioned for high visibility and clarity.
Rotating bobbin flowmeters
nicely calibrated for accuracy and clarity.
Plenum type Ether unit and
Halothene vapouriser fitted.
Nonreturn valve: to minimize
risk of backflow of gases.
Pressure relief valve: does
not allow pressure to exceed 200 cm of water
Top tray for
Box for storage
Self sealing valves – for oxygen
service for ventilators
Platform to house
two 10 liter water capacity cylinders
The photograph shows certain accessories, which
1. Circle Absorber 2 Cylinders. 3. Conversion
Kit for O2 & N2O
We reserve the right to change the design of Unit
without any notice.
During anaesthesia, the
expired gases still contain some of the anaesthetic
gases and this may be rebreathed if the carrbondioxide
is removed in the absorber.It consists of a container
filled with sodalime through which the patient’s
expired gases are passed.
This is efficient in the use of anaesthetic gases,
reduces pollution of the room air, conserves heat
and moisture and does not normally require humdification.